24V noiseless DC motor chair 1 set
Auto spittoon flush and cup filler control system 1 set
Luxury glass spittoon(detachable) 1 set
Luxury LED light 1 set
3-way syringe(Cold/warm water) 2 sets
Saliva ejector with high and low suction 1 set
Floor box with main switch(Air/water) 1 set
Removable instrument tray 1 set
Imported electromagnetism valve 1 set
LED x-ray film viewer 1 set
Dentist stool C 1 set
Power Voltage 110V/220V-50HZ/60HZ
Motor Voltage 24V
Air Pressure 0.5MPa-0.8MPa
Water Pressure 0.2MPa-0.4MPa
1）In order to avoid blocking of the internal tubing and valve for dental unit, you should use clean water.
2）You should pay attention to draining sewage, there is moisture in the air, according to the local air humidity, you should drain dirty water one month once(if it is large load work, it should be done half month)
3）Dental chair is the integration stage between machine and motor, so you should lubricate the motor after using half year.
4) Try to avoid lacerating the cushion by key or other sharp things from patients.
5) When the power supply is not stable, do not use dental chair, in case damage electronic parts inside of dental chair.
6) In summer, unplug the water heater in the box (you could consult your local distributor). When the temperature below zero in winter, and the same time you are using dental chair, please blow inside water to be dry, otherwise the water heater may freeze to be damaged under very low temperature during night.
7) After one day of work, turn off the power, water and air switches.